Normalization plate kit for the TopCount liquid scintillation counter. This 96-well SPC (single photon counting) plate is used for TopCount luminescence normalization (for the 2, 6, or 12 detector system). The plate itself is a LumaPlate (Yttrium silicate, solid scintillator), with 3H thymidine impregnated epoxy.
Item(s) added to cart.
Please enter valid quantity
Please log in to add favorites.
NULL OR EMPTY CART
Each time you create a new assay you must normalize the instrument’s detectors. This is called per assay normalization. You should normalize the instrument’s detectors before you use the newly created assay to count samples.
During the normalization procedure, the system calculates a factor for each detector that compensates for counting differences between detectors. When the system applies the factor to the value measured by the detector, the final result is that different detectors counting the same sample will yield similar CPM and tSIS values. This balances the responses of the detectors against one another.
|Special Ordering Information||This is a radioactive product - shipping address must have a license to receive radioactive materials.|
For over 50 years PerkinElmer has been a leading supplier of radiochemicals, liquid scintillation cocktails, vials and nuclear counting detection instruments. Today is no different. We have always been committed to providing you products for all of your radiometric needs and we are still committed today.